We are offline at the moment
Leave a message
View My Virtual Trade Show Booth
Charles Hill, Operations Manager
DOTmed user since September 2014
Item has been used previously and shows signs of cosmetic wear. Unit powers on and appears to function as intended. No further testing can be conducted. Item being sold...
MPN: HF 300...