Philips BV Endura error M325
June 17, 2019 05:11
Can anyone help me please with the error M325
to rate and post a comment
re: Philips BV Endura error M325
June 19, 2019 09:19
X-ray generator not available. Switch system off/on. Call service if problem is persistent.
Logfile text: The self test of the FA Generator Control Board SHA4 has failed. The X-ray function is disabled.
Check the cabling between the Generator Control Board and the SE unit. Replace the Generator Control Board.
15,000 service professionals from over 50 different countries have trained in over 35 courses in radiology, mammography, CT, ultrasound, networking, PACS, & DICOM. Classroom & lab facilities to accommodate up to 60 students.