May 13, 2019 10:39
Hello i have problem with oec 9900 at boot touch-screen don't working when i touch it and of tra to fluoro after fluoro COMUNICATION error somebody can help me about this problem?
to rate and post a comment
May 15, 2019 10:25
I would start by checking the usb cable and connections from the touch screen monitor to the GPOS CPU. GPOS is the cpu with the usb extender on it. the touch screen should be one of the cables plugged into the cpu onboard usb jacks. I might be more helpfull if I have more details after you try this. firstname.lastname@example.org
Family owned since 1919, for your next medical move, rely on the specialists at Brandon Transfer & Storage. For references from the medical community & a free moving estimate, call us at 561-683-5000 and we care! Visit www.brandontransfer.com Today!