fuji cr-ir392 prima t2
March 01, 2017 10:04
I have fuji cr-ir392 prima t2 with a Fuji Console advance software(version 7.4) installed workstation.
My problem are :
1) the software is processing the image but after it i can't measure lenght of bone or the angle of pelvis and femur.No choice for measurement plugin.
2)I downloaded pacs software to examine the cr image but fuji keeps xray data images as *.std files and no pacs software view it.
3)when i choose softwares other installed programs ,it require password.But i dont have. (user name :guest but i dont have pass.)
4)Cr service manuel also will be appreciated.
to rate and post a comment
ROBERTO DIAZ BEDOLLA
re: fuji cr-ir392 prima t2
March 02, 2017 09:19
March 02, 2017 09:20
1) you may have the SOP User Classes might ne set up incorrectly in your DICOM Nodes or you may not have DICOM Send key installed. Fuji FDX Console Advance is NOT as such a viewing software with tools to measure. It is simply an aquisition software, limited to QA attributes. Yes, only a PACS or a Viewing software can do that. You cannot access the raw image folder from the Fuji Data Base, export it and expect to be able to view it on a secondary software. I am afraid that work around is difficult to succesfully achieve. Check your Options installed on FDX and you should have DICOM Store if not, then you cannot Send out. And if you do and still cannot send, then go back to DICOM Configurations.
re: re: fuji cr-ir392 prima t2
March 02, 2017 04:16
thank you for your kind reply
Jonas Ezra Gaerlan
June 09, 2019 04:46
Hi I have the same unit FUJI CR-IR392 prima T2 with Fuji Console Advance Software (version 7.02). But my problem is I'm stuck on the loading screen Fuji console advance, after 5-10mins it will terminate without errors or message. is this a software related problem? can you help me?
re: re: re: fuji cr-ir392 prima t2
June 23, 2019 03:03
you need to install the software again if you want i can help you to get the software
re: re: re: re: fuji cr-ir392 prima t2
February 12, 2020 09:50
Hi i need the sotware for the CR Prima CR-IR 392 give me the error 20102, i think the software lost the keys, any help?
re: re: re: re: re: fuji cr-ir392 prima t2
February 13, 2020 09:27
I checked your error code from service manual.It writes down;application key software key is not installed.
March 02, 2017 09:21
Reinstal aplication software
This solves nothing.
March 03, 2017 09:33
He cant measure -it mean problem with SW
March 06, 2017 08:50
Fuji acquisition software doesn't come with measuring tools. No, is not a software problem, at least not a Fuji Software issue.
March 02, 2017 09:35
I have a NETPIX fuji console. I need to restore the ghost image of the system. I have ghost images. how to do?
You need acronics restore disc, start pc insert acronics disc then find img and restore
March 06, 2017 08:06
I tried with acronis but acronis doesnt accept the 3 image files.
March 12, 2017 10:29
it is ghost file not acronis
March 02, 2017 09:33
March 06, 2017 09:10
Write us to: email@example.com
Call Master Medical today for a great value on our large in-stock selection of
Defibrillators including Zoll, Physio Control, Philips with our Low Price Guarantee and Quick shipment. (866)468-9558.