Seller assumes all responsibility for this listing.
Condition: Used - Good
Software Version: 2.06
Always looking for Alcon Infiniti Phacoemulsifiers in any condition. We will fully PM and make patient ready.
We have the ability to supply parts for the ALCON INFINITI System. We also offer a full service course to the BIOMED so that you can fully maintain this system on your own. If you need parts, support or training, we can do it.
All the systems we sell are fully up to date, and patient ready.
1yr. Warranty on all systems
All variations, software versions, are wanted. We guarantee all systems, Preventive Maintenance, Quality Assure all systems before going out. Please call or email for availability and specifications. AND If you have any phaco system to sell, let us know.
This list of error code Alcon Infiniti is a partial list. If you need additional information, contact us and we will be happy to assist, just as we have trained alot of biomeds on the Alcon Infiniti Service and the Alcon Infiniti PM procedure. We will be happy to help you with parts and training.
THE LISTED ERROR CODES ARE SHOWN ON THE INFINITI VISION SYSTEM DISPLAY PANEL WHEN THE SYSTEM DETECTS A PROBLEM. THE ERROR CODES ARE SEPARATED BETWEEN ADVISORIES, WARNINGS, AND FAULTS.
Alcon Infiniti Error Code 160: Reinsert FMS. Replace FMS if Problem Persists.
Alcon Infiniti Error Code 161-164: Please Check Fittings and Reprime. Replace FMS if Problem Persists.
Alcon Infiniti Error Code 281: Cautery Compliance Error. If Problem Persists After Restarting System, Contact Alcon Technical Services.
Alcon Infiniti Error Code 282: Coagulator Error. If Problem Persists After Restarting System, Contact Alcon Technical Services.
Alcon Infiniti Error Code 350: Footswitch Failure Detected. Check Footswitch, Clean Under Rear Section of Treadle, and Remove Debris if Present. (Reference Maintenance Section of Operator’s Manual.) Ensure Treadle is Not Depressed, then Reset Footswitch. If Condition Persists, Contact Alcon Technical Services.
Alcon Infiniti Error Code 351: Footswitch Failure Detected. Check and Reset Footswitch. If Condition Persists, Contact Alcon Technical Services.