Here is a Zeiss Axiovert 200M Fluorescence Motorized Phase Contrast Inverted Microscope Pred AXIO Observer
Setup for Brightfield with Phase Contrast and Fluorescence
This Zeiss Axiovert 200M Fluorescence Motorized Phase Contrast Inverted Microscope is in good working order with all aspects tested and working fully.
Microscope and optics have been professionally cleaned by our technicians under a class 10,000 clean room hood.
Please see the exact video and photos below.
Testing we performed:
The Zeiss Axiovert 200M Fluorescence Motorized Phase Contrast Inverted Microscopepassed all tests and was found to be fully functional:
Tested initial power to the microscope: PASSED
Tested metal halide lamp and adjustment: PASSED
Tested halogen lamp and adjustment: PASSED
Tested camera and binocular ports: PASSED
Tested Z-axis coarse and fine focus: PASSED
Tested mechanical XY stage: PASSED
Tested motorized Z-axis: PASSED
Tested motorized fluorescence turret: PASSED
Tested motorized condenser turret: PASSED
Tested brightfield light path: PASSED
Tested fluorescence light path: PASSED
Tested objective for clarity: PASSED
Tested filters and prisms: PASSED
Tested MU1003 camera live feed and clarity: PASSED
Tested all buttons functionality: PASSED
During testing this Zeiss Axiovert 200M Fluorescence Motorized Phase Contrast Inverted Microscopewas found to have no issues.
A clear image was produced through the eyepieces as well as through the camera port.
All of the mechanical and motorized components moved freely and the test sample was able to be focused without excessive effort.
System consists of:
Zeiss Axiovert 200M Inverted Microscope
(2) W-PI 10x/23 Eyepieces
Prisms / Condensers:
Motorized 6-position Condenser NA:0.55 WD:26mm
Brightfield Light Path:
Zeiss Illumination Arm
Fluorescence Light Path:
Zeiss Fluorescence Light Path
Fluorescence / Cubes:
Motorized 5-position Filter Cube Turret
CY3-EX:535/50nm DM:565nm EM:610/75nm
B-1E- EX:470-490nm DM:505nm EM:520-560nm
GFP BP-EX:480/40nm DM:505nm EM:535/50nm
HAL100 12V 50W Halogen Lamp w/ built-in power supply
EXFO X-Cite 120 Metal Halide Lamp w/ light guide
Mechanical XY Stage
Custom Stage Insert
Motorized 6-position Objective Turret
20x/0.60 Plan-APOCHROMAT Ph2
40x/1.30 Plan-NEOFLUAR Oil
Objective Turret, Fluorescence Cube Turret, Z-axis, Condenser Turret, Fluorescence Shutter and light path switching
AmScope MU1003 10MP CMOS
Computer and Software:
Dell Latitude i5 running Windows 7 Pro with AmScope software
Communication and power cables.
Configuration and Setup:
Illumination: Brightfield with Phase Contrast and Fluorescence
Objectives: 10x/0.25 CP-ACHROMAT, 20x/0.60 Plan-APOCHROMAT Ph2, 40x/1.30 Plan-NEOFLUAR Oil
Light Sources: 12V 100W Halogen Lamp, X-Cite 120 Metal Halide
Stage Type: Mechanical XY w/ Motorized Z-axis
The Zeiss Axiovert 200M Fluorescence Motorized Phase Contrast Inverted Microscopeis currently set up for Brightfield viewing with Fluorescence and Phase Contrast studies.
The AmScope CMOS camera is capable of taking detailed digital images. The system includes a Dell Latitude i5 laptop to control AmScope software.
With included 10x,20x,40x objectives and 10x eypieces the total magnification of the system is 400x.
This video begins with a full front view of the Zeiss Axiovert 200M Fluorescence Motorized Phase Contrast Inverted Microscopeand all components powered up and running.
The camera scans from one side to the other for a closer look. Next, we see the AmScope live feed, focus, stage XY, and light intensity, objective selection, and filter cube.
TO VIEW ADDITIONAL PHOTOS, VIDEOS AND VERY DETAILED INFORMATION SUCH AS TEST STEPS PERFORMED, CONFIGURATION, ETC., PLEASE VISIT OUR WEB SITE. TO GET THERE PLEASE CLICK ON "WEBSTORE" TO YOUR RIGHT AND YOU WILL SEE LINK TO OUR SITE IN THE MIDDLE OF THE SCREEN. THEN USE LAST 5 DIGITS OF MPN NUMBER IN THE SEARCH FIELD TO LOCATE YOUR ITEM.
DO NOT HESITATE TO CALL US OR EMAIL US WITH ANY QUESTIONS YOU MIGHT HAVE.
ALL OF OUR ITEMS ARE AVAILABLE FOR INSPECTION AT OUR FACILITY.
Items are sold as-is with no returns or refunds available unless explicitly stated.