BioDot CSTM AD1510 Liquid Handler Pred. to AD1520 Unit2
Made in 2012
Current model is AD1520
This BioDot CSTM AD1510 Liquid Handler is in good working order and also has good cosmetic condition.
This BioDot CSTM AD1510 Liquid Handler came from a project that was canceled before this system was setup and is unused.
Please see the exact video and photos below.
Testing we performed:
The BioDot CSTM AD1500 Liquid Handler passed all tests and was found to be fully functional:
Communication with control software: PASSED
Stage movement XYZ: PASSED
Syringe pump movement: PASSED
Wash Pumps: PASSED
Shuttle 1 and 2 movement: PASSED
Door Sensors: PASSED
System consists of:
BioDot AD1500 CSTM Aspirating and Dispensing Platform
Syringe Pump with 250ul Syringe
BioJet Plus Fixed Tip
(2) Shuttle Controllers
(2) Nest Shuttles (Nest Attachments Not Included)
ILD5000 4-channel Stand Alone Vacuum Degasser
Holder Installation Tool for BIOJET Plus
Degasser and Sample Vials
Moisture Sensitive Slides
Dell Latitude i5 running Windows 7 Pro with AxSys software ver2
Communication and power cables.
Configuration and Setup:
Item Type: Liquid Handling System
Dispense Tips: Fixed BioJet Plus Single Channel
Dispense Precision: <10% at 20nl
XY Position Accuracy with Shuttle Nest: 0.025 mm
In Line Degasser: Included
This BioDot CSTM AD1510 Liquid Handler is capable of 250ul transfers using the BioJet Plus Fixed Tip of both chemical and biological reagents.
With the addition of substrate nests this system will be able to house 14 slides, 3 microplates, or membrane hold downs.
This system would be an excellent choice for Biosensors, Biochips, Low volume PCR, Multiplexing ELISAs and DNA or Protein Dispensing applications.
The video begins with a full BioDot CSTM AD1510 Liquid Handler view from the front.
The camera then pans in to show the control software running on the laptop.
The software is then restarted and the camera pans over to show the gantry perform its homing sequence.
The syringe pump is also shown to demonstrate operation. The camera pans back to the computer and the internal pumps for wash and waste are activated.
Next shuttles 1 and 2 are ejected one at a time with manual control.
A brief virtual tour of the system shows the back of the BioDot CSTM AD1510 Liquid Handler and the video ends with a final full system view.
TO VIEW ADDITIONAL PHOTOS, VIDEOS AND VERY DETAILED INFORMATION SUCH AS TEST STEPS PERFORMED, CONFIGURATION, ETC., PLEASE VISIT OUR WEB SITE. TO GET THERE PLEASE CLICK ON "WEBSTORE" TO YOUR RIGHT AND YOU WILL SEE LINK TO OUR SITE IN THE MIDDLE OF THE SCREEN. THEN USE LAST 5 DIGITS OF MPN NUMBER IN THE SEARCH FIELD TO LOCATE YOUR ITEM.
DO NOT HESITATE TO CALL US OR EMAIL US WITH ANY QUESTIONS YOU MIGHT HAVE.
ALL OF OUR ITEMS ARE AVAILABLE FOR INSPECTION AT OUR FACILITY.
Items are sold as-is with no returns or refunds available unless explicitly stated.